The reading depends on the plug-in element being used.
For example, if you have a 100 W element, you use the corresponding scale that ends in 100 (the bottom numbers on the display).
Similarly if you have a 10W element, you also use the same scale as the 100 W element but just move the decimal point (100 W to 10.0 W) for the correct reading. A 1W element also uses the same scale but move the decimal point two places to the left (100W to 1.00W).
If you have a 5W, 50W or 500W element you would use the scale that ends in 50 and if you have a 25W, or 250W element use the top scale.
Please leave us a comment on this article. Did it help you? What else should we include to make this article better? Thanks in advance from the Bird Applications Team!
Please sign in to leave a comment.