The meter’s zero setting should be checked when no RF power is present. When
no power is applied the pointer should rest exactly on zero. If adjustment is
required, turn the adjustment screw until the pointer is set at zero.
Peak power may need calibration. Please refer to the instruction manual for 43P Peak Power Calibration procedures.